Trying to understand readings expected for wiring a 10V output to a +/-10V input when calibrating on a B3000 Mistic mode. Firmware chips just changed to R81.36.
If I set output to 0 counts, and send ‘calculate and set offset’, then set output to 4095 and send ‘calculate and set gain’, then read the AIV module current counts, a 4095 on the output is 24999 on the input, and 0 on the output is near 0 on the input.
I would have expected the module to read 0 at 0V and 4095 at 10V when I use my AOV-25 to generate 0V and 10V and calibrate offset at 0V and gain at 10V.
Does an offset calculation expect 0V or -10V on the AIV +/-10V input for low scale?